|Usage:||Strength Testing Machine|
|Place of Origin:||China (Mainland)|
|chair test machine:||durability test|
|strength test:||furniture performance test|
Office Chair Seat Back durability tester is used for testing the strength and durability of office chair. Made of Aluminum material frame, PLC contral, easy operation; high precision force sensor for the load application.
Seat and chair back durability combined tester.
Seat and back Staic load test: test consisting of heavy loads being applied a few times to ensure that furniture has sufficient strength under the highest levels of loading that might reasonably be expected to occur.
Seat and back Fatigue test:Test simulating the repeated application of loads or movement of components occurring during long-term functional use.
This test appratus do the durability test to chair seat and backrest by simulating the loading application to the chair as daily use.First use the seat loading pad to apply the certain loading to the seat surface, then the back loading pad to the backrest, then loading unload to finish one cycle of test, repeat the loading movment some times and certian cycles according to different test requirement, to determin the durability permance of chairs.
Loading Sensor: 2500N (2nos) and 1000N (2nos) (or specified by customer)
Load force solution: 1/10000
System test force accuracy: ±1%(static) ; ±5%(dynamic)
Test speed: 10~30 cycles per minute (adjustable)
Loading cycle: 0~999999 adjustable
Power: 1∮AC 220V 50Hz 5A,3KW
Accessories: Seat loading pad 1 set; Back loading pad 1set; Stationary fixture 1 set.
Standards: BS EN 581-2 B.2.1.1 and B.2.1.2,EN 1728; BIFMA X5.1,QB/T2280 /GB 10357.3; and BS EN 1335.